This model combines the advantages of similar products found at home and abroad; it eliminates the shortcomings of the on-frequency X-ray machine (high current intensity and excess amounts of scrap x-rays). Quency of this machine is 30 KHz. And the tube current is 1 MA.
The radiation scope is at an angle of 24 degrees. Located within a distance of 1.2 meters ahead.This is an innovative Chinese design named the Green X-ray Machine. Focal Spot To Skin Distance. Reference of the time of exposure. Operation of the Main Unit. Make sure the fittings on the encasement list are packaged within the box. The pilot lamp will be light up. During this timethe digital tube will show the fore se. Now the equipment is in standby mode. Ing the time (Skip to next step if unneeded). "+"and" -" buttons to reset the time as follows: press SETthe pilot lamp above is illuminated the machine is now in the time setting mode. Then press "+" and" -" to set the time needed time range is. Seconds; The equipment is now in standby mode.
Put the Tooth film plumb behind the tooth which is going to be taken picture and be as close as possible the smooth side stick to the t. Keep the ball head plumb to the tooth projection position; Have the ball headtooth and tooth film steadily mutually plumb.After positioninguse the ON/OFF on the main Unit to take pictures. (Notice: gently press 0.5 sec to turn it on). Make sure the angle of filmtoothand ball head are properly aligned when taking a picture.
Keep them steady until the last step of picturing. Making sure that there is no change in any of their positions. Remember to turn the power 0.
Ge the battery if t. He machine cannot work and the red lamp lignt. After turning the power onwait one minute before taking picturesthis allows the ray to provide a steady ou. The equipment switches into protection mode automatically when the voltage is incorrect.
If this occurs it will not be able to carry out the normal work of. ON/OFF button could stop the objection and then the machine goes back to the preparation mode. Use high quality tooth film and developing liquid to make clear pictures.
Handle the ball head gently while in useso as not to damage the delicate component. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period.
Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality. Accordance with the scope of the technical parameters. Please read the manual carefully before using. Warning: After turning on the machine, wait for 1 minute, the voltage can be stabilized before normal use. After each photo taking, you need to wait for the normal value to return before taking a normal photo.
The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. Statement:The sale of this item may be subject to regulation by the U.Wish you have a nice day! The item "Dental Portable Equipment On-Frequency X-Ray Machine BLX-5(8PLUS) Low Dose FDA" is in sale since Thursday, January 2, 2020. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment". The seller is "dentistworld" and is located in CA.
This item can be shipped to United States.