And the tube current is 1 MA. The radiation scope is at an angle of 24 degrees. Located within a distance of 1.2 meters ahead.
This is an innovative Chinese design named the Green X-ray Machine. Warning: After turning on the machine, wait for 1 minute, the voltage can be stabilized before normal use. After each photo, you have to wait 30 seconds before you can take a normal photo. Focal Spot To Skin Distance.
Reference of the time of exposure. Operation of the Main Unit.
Make sure the fittings on the encasement list are packaged within the box. The pilot lamp will be light up. The digital tube will show the fore setting time.Now the equipment is in standby mode. Setting the time (Skip to next step if unneeded) "+". And" -" buttons to reset the time as follows: press SET.
The pilot lamp above is illuminated. The machine is now in the time setting mode.
Then press "+" and" -" to set the time needed (time range is 0.1-1.8seconds); The equipment is now in standby mode. Put the Tooth film plumb behind the tooth which is going to be taken picture. And be as close as possible (the smooth side stick to the tooth).
Keep the ball head plumb to the tooth projection position; Have the ball head. And tooth film steadily mutually plumb. Use the ON/OFF on the main Unit.
(Notice: gently press 0.5 sec to turn it on). Make sure the angle of film. And ball head are properly aligned when taking a picture. Keep them steady until the last step of picturing; making sure that there is no change in any of their positions.
Remember to turn the power 0ff when work is done. Charge the battery if the machine cannot work and the red lamp lignt. After turning the power on.Wait one minute before taking pictures. This allows the ray to provide a steady output. The equipment switches into protection mode automatically when the voltage is incorrect. It will not be able to carry out the normal work of filming. The ON/OFF button could stop the objection and then the machine goes back to the preparation mode. Use high quality tooth film and developing liquid to make clear pictures. Handle the ball head gently while in use. So as not to damage the delicate component. Preventing the machine from taking regular pictures. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period.
Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality. Other requests accordance with the scope of the technical parameters. Please read the manual carefully before using.SR-8230, SR-8230S Portable X-Ray Unit. FDA Statement: The sale of this item may be subject to regulation by the U.
Food and Drug Administration and state and local regulatry agencies. If the item you received is damaged or defective, Please message to us immidately.
Our guarantee does not cover accidental damage or wear of product caused by customer. Thank you very much for your understanding!
We understand the concerns and frustrations you might have, and will try our best to resolve the isuues. Serivce is our #1 priority at BAISTRA! If you message us immediately after placing the order, we will make every effort to assist you with your change request.
We offer the best items at the best prices. We strive to exceed all your expectations and put a smile on your face. We stand behind our products and pride ourselves on OUTSTANDING customer service. The item "USA Dental Portable X-Ray Machine Unit Mobile X-Ray Equipment BLX-5(8Plus)" is in sale since Tuesday, March 23, 2021. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment".
The seller is "azdental2020" and is located in SHENZHEN. This item can be shipped to United States.