The tube voltage frequency of this machine is 30 KHz, and the tube current is 0.2 MA. The radiation scope is at an angle of 24 degrees, located within a distance of 1.2 meters ahead. Toshiba 0.3mm x 0.3mm micro-focus tube 3. No protective operation room is needed when installing the complete machine.Flexible adjustment of position and angle of handpiece 5. Compact in volume, light in weight, convenient for carrying 6. Time is adjustable, high contrast of picture, imaging in focus 7.
There are vacuum components and high-pressure parts inside of the machine, so for your safety and the maintenance of the machine 8. The tube head is equipped with excellent radiation-shield protection except the radial mouth, and tests have shown that the radial index is zero. Non-film inspection and storage can be realized when corresponding to a sensor. 20cm Tube voltage: 60KV Relative Humidity(RH).
Operation of the Main Unit. Make sure the fittings on the encasement list are packaged within the box. No protective operation room is needed when installing the complete machine 4.
This model combines the advantages of similar products found at home and abroad; it eliminates the shortcomings of the on-frequency X-ray machine (high current intensity and excess amounts of scrap x-rays). Using a Toshiba 0.3 micro-focus tube. The tube voltage frequency of this machine is 30 KHzand the tube current is 1 MA. The radiation scope is at an angle of 24 degrees located within a distance of 1.2 meters ahead.
This is an innovative Chinese design named the Green X-ray Machine. Focal Spot To Skin Distance. Reference of the time of exposure.Install the blue ray tube to the foreside of the ball headthen tighten it. Turn on the Powerthe pilot lamp will be light up. During this timethe digital tube will show the fore setting time. Now the equipment is in standby mode. Setting the time (Skip to next step if unneeded) Use the "SET""+"and" -" buttons to reset the time as follows: press SETthe pilot lamp above is illuminated the machine is now in the time setting modethen press "+" and" -" to set the time needed (time range is 1-9seconds); press set again when the time is adjusted the light lamp is off. The equipment is now in standby mode. Put the Tooth film plumb behind the tooth which is going to be taken picture and be as close as possible (the smooth side stick to the tooth). Keep the ball head plumb to the tooth projection position; Have the ball headtooth and tooth film steadily mutually plumb. After positioninguse the ON/OFF on the main Unit to take pictures.
(Notice: gently press 0.5 sec to turn it on). After exposure to the tubepress the start button again in case of an emergency to stop shooting and return to standby status. (At this point there was no output rayso the dental films can be used again).
Turn the power off when the work is finishedpress POWER making sure the pilot lamp is off. Don't keep the machine long in standby mode. Make sure the angle of filmtoothand ball head are properly aligned when taking a picture. Keep them steady until the last step of picturing; making sure that there is no change in any of their positions.Remember to turn the power 0ff when work is done. Charge the battery if the machine cannot work and the red lamp lignt. After turning the power onwait one minute before taking picturesthis allows the ray to provide a steady output. The equipment switches into protection mode automatically when the voltage is incorrect.
If this occurs it will not be able to carry out the normal work of filming. When taking picturesthe ON/OFF button could stop the objection and then the machine goes back to the preparation mode. Use high quality tooth film and developing liquid to make clear pictures. Handle the ball head gently while in useso as not to damage the delicate component.
We usually set the time between 3-8 seconds when taking picturesbut the actual time of ray output is 0.2-0.5 seconds. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period. Development of films should be performed at temperatures ranging between 23-25 degrees Celsius to ensure the image quality.Other requests accordance with the scope of the technical parameters. Please read the manual carefully before using. SHIPS FREE FROM THE USA TO THE USA!! No months-long wait from overseas! The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies.
We are not responsible for undeliverable/incorrect addresses. And we will work with you to resolve it to your satisfaction. We GUARANTEE a reasonable solution on all orders. The item "USA Portable Dental X-Ray Machine Unit X-Ray Equipment BLX-5/BLX-5(8PLUS)/LK-C27" is in sale since Friday, August 28, 2020. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\X-Ray Machines & Equipment\Other X-Ray Equipment".
The seller is "dentistlove" and is located in Fullerton, California. This item can be shipped to United States.