Digital X-ray Machine Portable X Ray System Equipment Portable RVG Sensor Dental. Portable Dental X-Ray Unit Machine Low Dose Digital Film Imaging System BLX-5. This model combines the advantages of similar products found at home and abroad; it eliminates the shortcomings of the on-frequency X-ray machine (high current intensity and excess amounts of scrap x-rays).The tube voltage frequency of this machine is 30 KHz, and the tube current is 0.1 MA. The radiation scope is at an angle of 24 degrees, located within a distance of 1.2 meters ahead. This is an innovative Chinese design named the Green X-ray Machine. All the components of this product were originally imported to achieve an acceptable tolerance required for the digital controls and the machine. By doing this, product perform ance is more stable and reliable.
Using a Toshiba 0.3 x 0.3 micro-focus tube, the image is clearer and more delicate. This machine uses components from well established suppliers, allowing us to deliver to our customers a product of high quality. There are films included for testing purpose and all the consumables can be replaced by other similar products. There are vacuum components and high-pressure parts inside of the machine, so for your safety and the maintenance of the machine, please do not open the machine. Service and maintenance should only be performed by a trained professional.The tube head is equipped with excellent radiation-shield protection except the radial mouth, and tests have shown that the radial index is zero. Non-film inspection and storage can be realized when corresponding to a sensor. Exposure time: 1-9 seconds optional. Storage conditions: 5-40 degrees Celsius, relative Humidity. Net weight: 5.5 kg.
Dimension: 47 x 42 x 19 cm. The Angle should be proper while taking a picture. The film should be put vertically behind the tooth to be taken a picture while the case is for the front teeth; and the film should be put transversely behind the tooth to be taken a picture while the case is for the back teeth.The patient should use forefinger to press the film behind the tooth gently. Kodak film as the sent samples are strongly recommended. Reference of the time of exposure. Operation of the Main Unit. Make sure the fittings on the encasement list are packaged within the box. Connect the six-core cable (of the Main Unit) to the joint of the ball head and tighten the screw; Install the blue ray tube to the foreside of the ball head, then tighten it. After the Main Unit and ball head are perfectly fit together, plug in the international standard electrical source, grounding is strongly recommended. Turn on the Power;the pilot lamp will be light up. During this time, the digital tube of the main unit will show the fore setting time. Now the equipment is in standby mode. Setting the time (Skip to next step if unneeded) Use the "SET", "+", and" -" buttons to reset the time as follows: press SET, the pilot lamp above is illuminated, the machine is now in the time setting mode? Then press "+" and" -" to set the time needed (time range is 1-9seconds); press set again when the time is adjusted the light lamp is off.
The equipment is now in standby mode. Put the Tooth film plumb behind the tooth which is going to be taken picture, and be as close as possible (the smooth side stick to the tooth). Keep the ball head plumb to the tooth projection position; Have the ball head, tooth, and tooth film steadily mutually plumb. After positioning, use the ON/OFF on the main Unit, the remote controller, or the button on the touch head tube to take pictures. The three methods have the same effect.
(Notice: gently press 0.5 sec to turn it on). After exposure to the tube, press the start button again (available in three ways) in case of an emergency to stop shooting and return to standby status. (At this point there was no output ray, so the dental films can be used again). Turn the power of the main unit off when the work is finished, press POWER, making sure the pilot lamp is off.
Don't keep the main unit long in standby mode. Picture the tooth following the right operating procedure of the X-ray Unit, then take the film out from the mouth. Use a syringe to extract 4-5 mi of imaging liquid. Transfuse the liquid into the bag, nip both the ends and the sides of the bag with thumbs and first fingers(knead the film).
Open the end of the bag about ten minutes after transfusing the liquid into the bag. Take the film out, put it into clear water, and rinse it.
Don't wipe the surface of the film with the surface to prevent damage to the image. Observe and diagnose after the film becomes dry. Feedback is very important to us. It is impossible to address issues if we do not know about them!
Faction is our ultimate goal. We work hard to earn your trust and future business. A If you receive a defective item, you may return it or ask for a discount. Due to time zones, our reply to your questions may not be immediately. If it's non-workdays, please kindly be patient and wait till Monday.
This item is in the category "Health & Beauty\Oral Care\Temporary Teeth & Supplies". The seller is "ktnparts" and is located in this country: US. This item can be shipped to United States.